While Ambilytics can be used as a data management tool, it is so much more. It uses machine learning to analyse, model, and identify trends and patterns in your data.
Ambilytics can integrate data from most instruments. Contact us with the details of your equipment, and we’ll confirm compatibility.
No, Ambilytics is designed to bring efficiencies on projects no matter their size. Request a demo and see how it would work for you.
All our equipment was designed with this possibility in mind. We can use 2G, 3G or 4G networks, LoraWAN, Satellite or NBIoT depending on requirements.
Due to power constraints, a Gasflux and be deployed with a water level setup or a dissolved gas probe – but not both.
VOCs can be measured with a PID using any lamp without any influence from methane / low C range alkanes so you should be okay with the PID. The TVOC concentrations unlikely to be interfered by concentrations from methane as these are not ionised / detectable using any of the lamps – methane has a ionisation energy of 12.6eV.
The water level loggers are vented and therefore the need to compensate the data.
We calibrate VOC units with isobutylene which has a response factor of 1. VOC PID readings are always taken as indicative readings and not absolute values. Absolute values of any constituent VOC gas would require a lab based test. Isobutlyene is used by sensor manufacturers and considered a solid middle ground to use.
50mm is standard but we can also install 35mm if required.
Yes, a Li-ion battery can be used to power it (max.28 day deployment with hourly sampling).
There is no set number or guidance which states the number of units required. The risk assessor will need to base this upon the Conceptual Site Model so that they can investigate / test the plausible contaminant linkages and provide sufficient information for their risk assessment. This is usually a targeted strategy, normally at receptor points (point of ingress into buildings) but also deploy in boreholes which intercept potential pathways and potential sources so a thorough understanding of the sites gas regime can be established.
We fit humidity filters to the feed line to prevent water ingress but we recommend that GasfluX units are only installed where the SWL is below 600mm and that there is a clear section of response zone to allow gas ingress from the ground.
Our team of experts are always available to answer your questions and make sure that your project is managed effectively.
Yes, Ambisense can provide our environmental risk assessment software to companies based in most locations around the world including the US and Canada.